|Year : 2016 | Volume
| Issue : 4 | Page : 117-120
Distortion of digital panoramic radiographs used for implant site assessment
Rayyan Abdulhamid Kayal
Department of Periodontics, Faculty of Dentistry, King Abdulaziz University, Jeddah 21481, Saudi Arabia
|Date of Web Publication||13-Oct-2016|
Rayyan Abdulhamid Kayal
Department of Periodontics, Faculty of Dentistry, King Abdulaziz University, P. O. Box 3738, Jeddah 21481
Source of Support: None, Conflict of Interest: None
Aims: This study is conducted to determine the amount of distortion of digital panoramic radiographs.
Materials and Methods: Panoramic radiographs of all patients who received dental implants in the years 2012 and 2013 were selected from the records at the faculty of dentistry, King Abdulaziz University. Radiographs were analyzed using the R4 Kodak Software for linear measurements of implants length and width. The measurements were compared to the actual size of the implant, and the amount of distortion was calculated.
Results: A total of 169 implants were analyzed. Horizontally, there was a statistically significant increase of 0.4 mm in width in the radiographic measurement compared to the actual size in the incisor region. Vertically, the sample overall exhibited a decrease by 0.4 mm compared to the actual size. Incisors had the highest difference with a decrease of 1.7 mm in the radiographic measurements compared to actual size. The highest distortion was found in the incisor region for both diameter and length (1.1 and 0.86), respectively.
Conclusion: Digital panoramic radiographs show minimal to no distortion. The highest distortion is found in the anterior area.
Keywords: Digital radiographs, distortion, implants, site preparation
|How to cite this article:|
Kayal RA. Distortion of digital panoramic radiographs used for implant site assessment. J Orthodont Sci 2016;5:117-20
| Introduction|| |
In dental practice, radiographs are considered an important tool for the assessment of bone architecture. Panoramic radiographs are widely used as standard radiographic examination tool when planning for implant therapy, especially for assessing vertical bone availability. ,, In recent years, advances in radiologic technology such as the introduction of digital radiographic imaging lead to increased efficiency of radiographic studies. They emit low-dose radiation and provide fairly accurate surveys. 
Distortion of radiographic images is one of the limitations that are well recognized in panoramic radiographs. , The amount of distortion depends on several factors including the distance between the patient and the film,  type of machine used,  and position of the object measured within the mandible.  This distortion must be considered during clinical application for proper diagnosis and treatment planning.
Determination of the available bone height is vital for choosing the appropriate implant length. Several studies indicate that this could be safely achieved using conventional and digital panoramic radiographs. ,,, Moreover, digital panoramic radiography allows adjustment of magnification to get 1:1 image visualization and the use of software-based measurement tools for more accurate measurements.  On the other hand, measuring available bone width is not possible using panoramic radiographs. Bone width can be determined clinically using bone-mapping techniques or by cone-beam computerized tomography (CBCT) scan which has been shown to be more accurate than panoramic radiographs.  Hence, careful radiographic and clinical measurements are essential to determine the size of the implant for the intended site.
The objective of this study was to calculate the extent of distortion of digital panoramic radiographs by measuring the dimension of dental implants used as radio-opaque reference objects.
| Materials and Methods|| |
Panoramic radiographs were selected from patients' records at the Faculty of Dentistry, King Abdulaziz University after obtaining approval from the Research Ethics Committee. The records of patients who received dental implants in the years 2012 and 2013 were examined. All records that contained acceptable postoperative digital panoramic radiographs obtained with Kodak 8000 (Hemel Hempstead, United Kingdom) digital panoramic unit were included in the study. This excluded radiographs with any artifact that affects the implant measurement.
The panoramic images were divided into three areas; molar region, premolar and canine region, and anterior region. All radiographs were analyzed by two trained dental interns using the R4 Kodak Software (Carestream Dental, Herts, UK) and were blinded to the actual size of the dental implants. Interexaminer reliability was conducted by comparing the measurements of both examiners. The R4 Kodak Software was used to perform linear measurements of the radiological implant length from the implant platform to the apex of the implant and from mesial surface to the distal surface at the neck of the implant for the radiological implant width. The actual size of the implant is determined from the digital file of the patient where practitioners write the size of the implant used.
Student's t-test was used to determine significant differences between mean measurements and actual size. Furthermore, the extent of distortion was calculated for each implant by dividing the implant's measured dimension (in mm) on the postoperative panoramic radiograph by the implants' actual size.
| Results|| |
A total of 169 implants from 92 radiographs were analyzed. Of these, 96 were placed in the maxilla and 72 implants were placed in the mandible. The number of implants placed in the incisor, canine/premolar, and molar region were 13, 77, and 79 implants, respectively. Implant diameters were 3.5 mm (36 implants), 4.1 mm (101 implants), and 5 mm (32 implants). The lengths were 10 mm (35 implants), 11.5 mm (66 implants), 13 mm (67 implants), and 15 mm (1 implant). There was no significant difference between measurements performed by both observers.
The diameter and length of dental implants were measured on digital panoramic radiographs and were compared to the actual sizes of these implants. Examining implants placed in all regions collectively revealed no statistically significant difference between the actual diameter, which represents the width, of the implant and the measured value [Table 1]. When each region was examined separately, different results were obtained. In the incisor region, the radiographic measurement was statistically significantly greater than the actual size by 0.4 mm (P < 0.05). When the length of the implant was examined, the radiographic measurements of the sample overall were decreased by 0.4 mm compared to the actual size (P < 0.05) [Table 2]. Divided by regions, incisors had the highest difference with a decrease of 1.7 mm in the radiographic measurements compared to actual size (P < 0.05) [Table 2]. This is followed by the canine/premolar with decrease of 0.4 mm in the radiographic measurements (P < 0.05) [Table 2]. On the other hand, there was no significant difference in the molar region.
|Table 1: Comparison between the actual implants diameter and diameter on the X-ray|
Click here to view
|Table 2: Comparison between the actual implants length and length on the X-ray|
Click here to view
Distortion was calculated and presented in [Table 3]. The highest distortion was found in the incisor region for both diameter and length. The diameter was 10.3% less on the radiograph compared to actual size, whereas the length was 13.7% higher on the radiographs compared to actual size. To find out whether the maxilla or mandible is more affected with distortion, the maxilla and mandible were looked at separately [Table 4]. There was no difference between actual size and radiographic measurement when it comes to diameter in both mandible and maxilla. On the other hand, there was 0.4 and 0.5 mm decrease (P < 0.05) in the radiographic image length compared to actual implant length in both maxilla and mandible, respectively.
|Table 4: Comparison between the actual implants size and size on the X-ray in both maxilla and mandible|
Click here to view
| Discussion|| |
Digital panoramic radiographs are a valuable adjunct to implant dentistry as a preoperative diagnostic and treatment planning aid. In many cases, these radiographs along with proper clinical examination can be sufficient to determine the size and position of implants if distortion is accounted for. This study was conducted to help determine the amount of distortion associated with modern digital radiographs and the accuracy of the analysis software. As to be expected, there was a considerable amount of distortion in the radiographs with a mean difference of up to 1.7 mm. This is less than what was found in earlier reports which showed 2.3 mm of distortion in panoramic radiographs. These studies used cadaver skulls and measured the actual size of the alveolar process. When steel balls were used, the distortion was reduced to 0.2 mm.  This suggests that distortion of objects within the alveolar process is less than distortion in the ridge which is something to be considered when studying distortion.
The amount of distortion differed from one area to another. The most distortion was found in the anterior area. Other studies also found that the anterior region had the most distortion. Kim et al. measured implant sizes in the mandible and found that the magnification is higher in the anterior region in comparison to other areas.  The difference in diameter might be due to the curvature in the premaxilla which cannot be represented by the X-ray machine. On the other hand, the difference in length, which was higher, is probably due to the angulations of the implants measured, which follows the apico-coronal accesses of the premaxilla and anterior mandible, in relation the floor when the occlusal plane is parallel to the floor. This could be attributed to the inclination of the premaxilla which has usually some degree of hyperdivergency and it is almost never perpendicular to the Frankfort horizontal while on the other hand the panoramic X-ray film is always perpendicular to the Frankfort horizontal. This difference in angulation causes the image to be shorter. This means that more caution should be applied when placing an implant in the anterior area to minimize the risk of injuring the floor of the nose.
In the posterior region, the amount of distortion is usually examined in the third molar region. , These studies found minimal to nonsignificant distortion in this region. In this study, minimal to no distortion was found in the posterior area. Nevertheless, the vertical dimension should be assessed very carefully, and adequate safety margin should be established, especially when working in proximity of vital structures such as the inferior alveolar canal and the maxillary sinus.
| Conclusion|| |
The results of this and other studies show that digital panoramic radiographs are reliable tools of preimplant assessment and treatment planning. Nevertheless, distortion should be accounted for, especially in anterior region. It has been shown that there is a difference in measurements between panoramic radiographs and CBCT, and CBCT is more accurate than both periapical and panoramic radiographs. Although the radiation exposure of standard digital panoramic radiography was found to be much less compared to CBCT, some cases require CBCT scan for more accurate measurements.  It is the clinician's responsibility to determine the need for a more extensive radiographic study for the best outcome with the least possible risk.
I would like to thank the intern office and the team of dental interns who helped in data gathering.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Vazquez L, Nizam Al Din Y, Christoph Belser U, Combescure C, Bernard JP. Reliability of the vertical magnification factor on panoramic radiographs: Clinical implications for posterior mandibular implants. Clin Oral Implants Res 2011;22:1420-5.
Vazquez L, Nizamaldin Y, Combescure C, Nedir R, Bischof M, Dohan Ehrenfest DM, et al.
Accuracy of vertical height measurements on direct digital panoramic radiographs using posterior mandibular implants and metal balls as reference objects. Dentomaxillofac Radiol 2013;42:20110429.
Vazquez L, Saulacic N, Belser U, Bernard JP. Efficacy of panoramic radiographs in the preoperative planning of posterior mandibular implants: A prospective clinical study of 1527 consecutively treated patients. Clin Oral Implants Res 2008;19:81-5.
Danforth RA, Clark DE. Effective dose from radiation absorbed during a panoramic examination with a new generation machine. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2000;89:236-43.
Ezoddini Ardakani F, Zangouie Booshehri M, Behniafar B. Evaluation of the distortion rate of panoramic and periapical radiographs in erupted third molar inclination. Iran J Radiol 2011;8:15-21.
Hayakawa Y, Wakoh M, Fujimori H, Ohta Y, Kuroyanagi K. Morphometric analysis of image distortion with rotational panoramic radiography. Bull Tokyo Dent Coll 1993;34:51-8.
Ladeira DB, Cruz AD, Almeida SM, Bóscolo FN. Influence of the intergonial distance on image distortion in panoramic radiographs. Dentomaxillofac Radiol 2012;41:417-21.
Laster WS, Ludlow JB, Bailey LJ, Hershey HG. Accuracy of measurements of mandibular anatomy and prediction of asymmetry in panoramic radiographic images. Dentomaxillofac Radiol 2005;34:343-9.
Devlin H, Yuan J. Object position and image magnification in dental panoramic radiography: A theoretical analysis. Dentomaxillofac Radiol 2013;42:29951683.
Reddy MS, Mayfield-Donahoo T, Vanderven FJ, Jeffcoat MK. A comparison of the diagnostic advantages of panoramic radiography and computed tomography scanning for placement of root form dental implants. Clin Oral Implants Res 1994;5:229-38.
Kim YK, Park JY, Kim SG, Kim JS, Kim JD. Magnification rate of digital panoramic radiographs and its effectiveness for pre-operative assessment of dental implants. Dentomaxillofac Radiol 2011;40:76-83.
Bolin A, Eliasson S. Panoramic and tomographic dimensional determinations for maxillary osseointegrated implants. Comparison of the morphologic information potential of two and three dimensional radiographic systems. Swed Dent J 1995;19:65-71.
Schropp L, Stavropoulos A, Gotfredsen E, Wenzel A. Calibration of radiographs by a reference metal ball affects preoperative selection of implant size. Clin Oral Investig 2009;13:375-81.
Abbassy MA, Sabban HM, Hassan AH, Zawawi KH. Evaluation of mini-implant sites in the posterior maxilla using traditional radiographs and cone-beam computed tomography. Saudi Med J 2015;36:1336-41.
Pertl L, Gashi-Cenkoglu B, Reichmann J, Jakse N, Pertl C. Preoperative assessment of the mandibular canal in implant surgery: Comparison of rotational panoramic radiography (OPG), computed tomography (CT) and cone beam computed tomography (CBCT) for preoperative assessment in implant surgery. Eur J Oral Implantol 2013;6:73-80.
Dudhia R, Monsour PA, Savage NW, Wilson RJ. Accuracy of angular measurements and assessment of distortion in the mandibular third molar region on panoramic radiographs. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2011;111:508-16.
[Table 1], [Table 2], [Table 3], [Table 4]
|This article has been cited by|
||Efficacy of deep convolutional neural network algorithm for the identification and classification of dental implant systems, using panoramic and periapical radiographs
| ||Jae-Hong Lee,Seong-Nyum Jeong |
| ||Medicine. 2020; 99(26): e20787 |
|[Pubmed] | [DOI]|
||Two-dimensional UTE overview imaging for dental application
| ||Kilian Stumpf,Elena Kaye,Jan Paul,Stefan Wundrak,John M. Pauly,Volker Rasche |
| ||Magnetic Resonance in Medicine. 2020; |
|[Pubmed] | [DOI]|
||Relationship between Radiographic Misfit and Clinical Outcomes in Immediately Loaded Complete-Arch Fixed Implant-Supported Prostheses in Edentulous Patients
| ||Robert W. Slauch,Avinash S. Bidra,Glenn J. Wolfinger,Chia-Ling Kuo |
| ||Journal of Prosthodontics. 2019; |
|[Pubmed] | [DOI]|
||A comparative study of cone-beam computed tomography and digital panoramic radiography for detecting pulp stones
| ||Melek Tassoker,Guldane Magat,Sevgi Sener |
| ||Imaging Science in Dentistry. 2018; 48(3): 201 |
|[Pubmed] | [DOI]|
||Evaluation of the relationship between sleep bruxism and pulpal calcifications in young women: A clinico-radiological study
| ||Melek Tassoker |
| ||Imaging Science in Dentistry. 2018; 48(4): 277 |
|[Pubmed] | [DOI]|
||Image quality assessment of pre-processed and post-processed digital panoramic radiographs in paediatric patients with mixed dentition
| ||Isti Rahayu Suryani,Natalia Salvo Villegas,Sohaib Shujaat,Annelore De Grauwe,Azhari Azhari,Suhardjo Sitam,Reinhilde Jacobs |
| ||Imaging Science in Dentistry. 2018; 48(4): 261 |
|[Pubmed] | [DOI]|